The company's S-1 lays the groundwork for what is widely expected to be one of the largest initial public offerings of the year, second only to Uber's IPO in May. It's also...Technologyread more
Fraud investigator Harry Markopolos' accusations extended beyond GE's management to actuaries, auditors and analysts who he claims overlooked billions in liabilities.Marketsread more
Trump's tweet comes a day after Apple put out a press release describing the money it spends on U.S.-based suppliers and vendors.Technologyread more
CNBC combed through Wall Street research to see which stocks are still a buy after their earnings reports.Marketsread more
President Donald Trump held a call on Wednesday with the CEOs of three major U.S. banks, according to people with knowledge of the situation.Marketsread more
Despite aggressive strides, Waymo needs one thing before their self-driving cars become a seriously useful transportation system: people. We talked to the ones closest to it.Technologyread more
Scientists say the smoke plumes, filled with megatons of tiny, harmful particles, could travel to other areas of the world and cause serious respiratory problems for people.Weather & Natural Disastersread more
Some Weight Watchers loyalists applaud Kurbo by WW. But nutritionists worry Kurbo promotes an unhealthy relationship with food during an especially impressionable time.Health and Scienceread more
Benefits from what President Trump called "the biggest reform of all time" to the tax code have dwindled to a faint breeze just 20 months after its enactment, writes John...Politicsread more
Epstein, 66, was found in his cell in Manhattan federal lockup Saturday morning and transferred to a nearby hospital, where he was subsequently pronounced dead.Politicsread more
Air travelers faced delays at U.S. airports on Friday afternoon after a computer issue snarled processing of international arrivals.Airlinesread more
The insulin pumps subject to the recall connect wirelessly to other insulin equipment, including glucose meters, a monitoring system and controls that pump insulin.
"The FDA is concerned that, due to cybersecurity vulnerabilities identified in the device, someone other than a patient, caregiver or health care provider could potentially connect wirelessly to a nearby MiniMed insulin pump and change the pump's settings. This could allow a person to over deliver insulin to a patient, leading to low blood sugar ... or to stop insulin delivery, leading to high blood sugar and diabetic ketoacidosis," the FDA notice says.
The MiniMed 508 pumps can't be updated to address security flaws in the device's firmware, according to the notice. The company is offering alternatives with "enhanced built-in security capabilities."
In a letter to patients, Medtronic urged customers to speak with their healthcare providers about whether to change the pump. For those continuing to use it, the company recommends they keep insulin pump and devices connected to it "within your control at all times" and advises customers not to share the pump's serial number, among other recommendations.
Medtronic has identified around 4,000 patients using the insulin pumps today, and is "working with distributor partners to identify additional patients potentially using the pumps."
A Medtronic spokesperson said Thursday evening the announcement was a "safety notice," and noted: "in the medical device industry, the term 'recall' is used generally to cover a range of actions including, for example, customer or patient communications with additional instructions for use of the product."
The spokesperson also said the company and FDA are not aware of any confirmed reports of a cyberattack on the pumps.
Medtronic's stock was steady Thursday.